1.Liquid barrier function: a) hydrostatic pressure of key parts shall not be less than 1.67kpa (17cmh20). b) The moisture permeability of the material shall not be less than 2500g / (m · d).
2.The anti synthetic blood penetration of protective clothing shall not be less than 1.75kpa.
3. The water level on the outer surface shall not be lower than the requirement of level 3.
4. Elongation at break: the elongation at break of materials at key parts shall not be less than 15%.
5. The filtration efficiency of key materials and joints for non oil particles shall not be less than 70%.
6. Antistatic property: the charged quantity shall not be more than 0.6 μ C / piece.
7. Electrostatic attenuation performance: the electrostatic attenuation time shall not exceed 0.5s.
8. Ethylene oxide residue: no more than 10 μ g / g.
Size: 30cm * 42cm * 44cm (length, width and height)
Specification: 20 sets / box
Scope of application of the product (there is scope of application on the registration certificate): it is used for blocking and protecting the potentially infectious patient's blood, body fluids, secretions, etc. which are contacted by the clinical medical staff at work.
Capacity: 10000 sets / day
Net weight / gross weight: 12kg (gross weight)
[structure composition] it is made of non-woven fabric sewing and bonding.
[scope of application] it is used for the barrier and protection of blood, body fluids, secretions and other potentially infectious patients contacted by clinical medical staff at work.
[usage] unfold the protective clothing, open the zipper, stretch into the legs from the zipper opening in turn, then put on the coat, stretch out the hand from the sleeve, wear the headgear, and pull the zipper from the front of the chest.